The BG-Malaria trap was recently modified from the BioGents BG-Sentinel trap to collect Anopheles species, including Anopheles darlingi. However, the captured mosquitoes often lose their hind legs in the collector bag, making them difficult to identify. To develop a new collector system that is capable of maintaining the integrity of the mosquitoes collected in the BG-Malaria trap, we conducted a study in the municipalities of Belém (Pará State [PA]) and Porto Velho (Rondônia State [RO]), Brazil, using carbon dioxide–baited BG-Malaria traps with 4 different mosquito collector systems: standard, no bag, rigid short, and rigid long. Results indicated significant differences among the numbers of mosquitoes captured in the 4 different collectors (P < 0.05). Additionally, significantly fewer insects (P < 0.05) were damaged using the rigid short and rigid long collectors than by using the standard and no-bag collectors. We observed that the longer the insects remained in the collector, the higher the number of damaged insects; this effect was the greatest in the standard collector. The results of this study indicate that rigid long collectors were the best suited for use in the BG-Malaria trap.
You have requested a machine translation of selected content from our databases. This functionality is provided solely for your convenience and is in no way intended to replace human translation. Neither BioOne nor the owners and publishers of the content make, and they explicitly disclaim, any express or implied representations or warranties of any kind, including, without limitation, representations and warranties as to the functionality of the translation feature or the accuracy or completeness of the translations.
Translations are not retained in our system. Your use of this feature and the translations is subject to all use restrictions contained in the Terms and Conditions of Use of the BioOne website.