The performance of two light-emitting diode traps with white and black light for capturing phlebotomine sand flies, developed by the Argentinean Leishmaniasis Research Network (REDILA-WL and REDILA-BL traps), were compared with the traditional CDC incandescent light trap. Entomological data were obtained from six sand fly surveys conducted in Argentina in different environments. Data analyses were conducted for the presence and the abundance of Lutzomyia longipalpis, Migonemyia migonei, and Nyssomyia whitmani (106 sites). No differences were found in presence/absence among the three types of traps for all sand fly species (p>0.05). The collection mean of Lu. longipalpis from the REDILA-BL didn´t differ from the CDC trap means, nor were differences seen between the REDILA-WL and the CDC trap collection means (p>0.05), but collections were larger from the REDILA-BL trap compared to the REDILA-WL trap (p<0.05). For Mg. migonei and Ny. whitmani, no differences were found among the three types of traps in the number of individuals captured (p>0.05). These results suggest that both REDILA traps could be used as an alternative capture tool to the original CDC trap for surveillance of these species, and that the REDILA-BL will also allow a comparable estimation of the abundance of these flies to the CDC light trap captures. In addition, the REDILA-BL has better performance than the REDILA-WL, at least for Lu. longipalpis.
You have requested a machine translation of selected content from our databases. This functionality is provided solely for your convenience and is in no way intended to replace human translation. Neither BioOne nor the owners and publishers of the content make, and they explicitly disclaim, any express or implied representations or warranties of any kind, including, without limitation, representations and warranties as to the functionality of the translation feature or the accuracy or completeness of the translations.
Translations are not retained in our system. Your use of this feature and the translations is subject to all use restrictions contained in the Terms and Conditions of Use of the BioOne website.
Vol. 40 • No. 2